© 2018. RCB Chemicals & Services. All rights reserved.
The industrial standard KE- ORPS 1001 sensor is a solid dielectric sensor used for continuous monitoring of Oxidation Reduction Potential of surface water, ground water, alkali chemicals, chlorination systems, pigments, dyes, pulp and paper, electroplating industries and industrial effluent water. The durable PTFE body has an excellent chemical compatibility with anticlogging design which provides easy maintenance.
1- Long Distance reference diffusion path extends electrode life even in harsh environments.
2- High accuracy with fast response and good repeatability.
3- Electrode made of high quality low noise cable, which provides signal output without any interference.
4- Pt1000 RTD, the temperature element used in the KE- ORPS 1001 sensor is highly accurate and linear throughout the range.
|Measuring Range (ORP)||Upto ±2000mV|
|Maximum Operating Temperature||85.0°C|
|Safety Pressure||6 Bar|
|Connection Mode||Tinned Wire Connection|
|Cable Length||5 Meter|
1- Do not put Sensor directly into water, the sensor should be installed with Female Thread Socket for Inline Flow through Measurement and Threaded Sensor Housing for Offline Measurement.
2- Before installation, use Teflon Tape (at 3/4" Thread) to ensure no leakage occurs
3- Connect the Sensor to Controller, let it stable for 3 Mins, then start calibration work.
4- Connect / Insert the Sensor vertically upright in the Solution whenever doing calibration.
5- Sensor to be installed Inline to piping where the pipe line doesn’t get dry whenever process is not operational.
Image above shows the wiring connection names for RCB ORP Sensor KE-ORPS 1001 and RCB ORP Controller KE-ORP 1000. Each Sensor wire connection detail is given below:
1- For 2 wire temperature sensor, short T/P4 and T/P3. Connect TEMP Wire 1 (Pink) with T/P 1 and TEMP Wire 2 (Black) with T/P 2.
2- Connect INPUT Wire 3 (Copper) with GLASS connection of Controller.
3- GND Wire 4 (Green) can be used if required.
4- Connect REF Wire 5 (Yellow) with REF connection of Controller.
1- In case of any issue other than mentioned above, immediately inform RCB representative about the issue which will be resolved at the earliest.
2- Special fittings Inline or Offline for ORP Sensor KE-ORPS 1001 can be ordered to RCB separately.