© 2018. RCB Chemicals & Services. All rights reserved.
The industrial standard KE- CDS 2000 series sensor is used for continuous monitoring of electrical conductivity / TDS of surface water, ground water, pure water, aqueous solutions and industrial effluent water. The durable PTFE enclosure consisting of 316L Stainless Steel twin electrodes which have an excellent chemical compatibility with anticlogging design which provides easy maintenance.
1- High accuracy with fast response and good repeatability.
2- High level of process safety due to durable materials and robust design.
3- Widely spaced electrode which do not trap ion exchange resin particles, corrosion particles or bubbles and yield more reliable, maintenance-free operation.
4- Pt1000 RTD, the temperature element used in the KE- CDS 2000 Series sensor is highly accurate and linear throughout the range.
|0.01||0-20 μS/cm||5 Bar||SS316L/Titanium Alloy||PTFE||3/4” BSPT||Power Plant & Water Treatment||CDS2001|
|0.1||0-200 μS/cm||5 Bar||SS316L/Titanium Alloy||PTFE||3/4” BSPT||Power Plant & Water Treatment||CDS2002|
|1.0||0-2000 μS/cm||5 Bar||SS316L/Titanium Alloy||PTFE||3/4” BSPT||Water Treatment||CDS2003|
1- Do not put Sensor directly into water, the sensor should be installed in Female Thread Socket for Inline Flow through Measurement and Threaded Sensor Housing for Offline Measurement.
2- Before installation, use Teflon Tape ( 3/4" Thread) to ensure no leakage occurs.
3- Connect the Sensor to Controller, let it stable for 3 Mins, then start calibration work.
4- Connect / Insert the Sensor vertically upright in the calibration Solution whenever doing calibration.
5- Sensor to be installed Inline to piping where the pipe line doesn’t get dry whenever process is not operational.
6- Sensor to be installed in pipe lines opposite to flow direction.
Note: All Dimension are in Millimeters unless otherwise mentioned.
Above image shows the wiring connection names for Conductivity Sensor KE-CDS 2000 series and Conductivity Controller KE-CD 1000. Each Sensor wire connection detail is given below:
1- For 2 wire temperature sensor, short T/P4 and T/P3. Connect TEMP Wire 1 (Yellow) with T/P 1 and TEMP Wire 2 (White) with T/P 2.
2- Connect DD Wire 3 (Black) with CN3 connection of controller and short CN3 with CN4.
3- Connect DD Wire 4 (Red) with CN2 connection of controller and short CN2 with CN1.
1- In case of any issue other than mentioned, immediately inform RCB representative about the issue which will be resolved at the earliest.
2- Special fittings inline or offline for conductivity sensor KE-CDS 2000 series can be ordered to RCB separately.
3- Threaded connection can be changed to MNPT standard on user’s request.